This automatic optical fiber tensile testing system is ideal for operator skill impartial use within creation and laboratory configurations. Two fundamental design designs enable the user to choose between a set up for regular linear evidence testing with additional mandrels for rotary high strength testing.
The linear edition is designed for creation evidence testing with forces approximately 22 N. Motor-driven clamps with carefully developed rubber inserts be sure that the Optical Fiber Proof-Testing Machine is not really ruined within the screening procedure while at the same time eliminating the need for an external air provide. Degrees of pulling force, pulling speed and hold time are programmable and easily accessible to the operator with the built-in LCD touch-screen manage. Data from tests can be saved and reached using the Universal serial bus port or Ethernet user interface.
Mandrels of the identical style like NYFORS’ optical fiber recoaters can easily be installed on the evidence tester within the rotary higher power configuration. The fiber is wrapped around the two mandrels which support it having a special clamping mechanism. By doing this higher lots can be employed on the fiber, enabling breaking strength tests below managed circumstances.
Changeable clamp positioning facilitates testing with the highest force amounts as well as testing of brief fiber measures. The lightweight and light bodyweight style makes the instrument simple to maneuver in a lab environment and to integrate in a creation work table where space reaches a premium. The equipment is actually a compact and light-weight tool perfect for testing the potency of optical materials as much as 30 N. The equipment is created to fulfill all sorts of applications when a power check must be carried out.
The equipment was created using the consumer in mind. The SZ Stranding Line is definitely put into the carefully guided linear clamps, as well as the screening starts instantly having a “One Touch” go switch. The built-in microprocessor controls all of the essential parameters such as pulling force, tugging velocity and hold time. This permits for any totally controlled test every time.
The equipment can be connected to an external Computer, which allows use of all programmable parameters and settings. The equipment is tailored to face-on your own use or even to the incorporation into various subsystems. The device posseses an extensive capability of logging and saving details including pulling force, pulling speed and fiber size as well as other parameters.
Two recoat injector configurations can be found. The PTR307 uses an automated pump to inject the recoat material. The quantity of material dispensed from the automated injector is controlled by hand through the top-installed “Inject” button or programmed to the machine using the tablet control. The PTR307B features a manual recoat shot system that requires an individual to personally dispense the recoat materials in to the mold cavity; another one injector for that PTR307B recoater is accessible below. Please note that the automatic injector is just appropriate for higher-directory recoat materials, while the manual injector is compatible with both low- and high- index recoat material; both are marketed individually listed below. Please get in touch with Tech Assistance to learn more.
he manual injector can be mounted to compatible SZ Stranding Line using the 4-40 screws on the recoater real estate (see photo towards the right). Use a 3/32″ hex key to safe the injector before use. To get in touch the PTRRRM for the recoater mold, tighten up the connector at the end of the green plastic tubing, then loosen by a 1/4 choose permit rotation.
The injector is equipped with a syndication device and two-place choice handle for directing the flow of recoat material. A knurled dispensing attach with the internal plunger behaves as a syringe for that recoat material. To fill the syringe, point the handle downward (i.e., towards the recoat container), then rotate the knurled dispensing attach counterclockwise till it spins easily to fill up the syringe (demonstrated in the photo for the right). Then, to inject the recoat materials to the mold, point the lever horizontally (i.e., facing the knurled screw) and rotate the attach clockwise till close to the finish from the travel range is reached. Avoid bottoming out iuwekk dispenser as this may harm the inner plunger; also be careful when re-engaging the threads to prevent go across threading the dispensing screw. Several fill/inject actions may be needed until air is displaced in the system. Use camera lens cells plus an acetone or alcohol cleansing solution to gather any excess recoat material that flows from the mold.